All-Sun EM2271 tester is designed to not-destructively measure the thickness of non-magnetic coatings (including paint enamel ,plastic ,epoxy, etc) on iron ,steel or Aluminum.
All-Sun EM2271 SPECIFICATION:
Operating Environment: Temperature: 0-40°C
Temperature Coefficent: 0.05x(2%rdg±0.01mm,or 2%rdg±4mil)/ °C
(<18 °C or >28 °C )
Storage Environment: Temperature:-10 °C to 50 °C
All-Sun EM2271 Operating Instruction:
1. Hold the tester and make sure that it is as least 20cm away from the
object to be tested and other objects.
Then press the "TEST"button to turn on the tester. " -- " appear on the
display and flashed,meanwhile the unit "mm"appear.
2. Press the "UNIT"button to select desired unit - mm or mil.
3. Place the tester's probe perpendicularly on the surface to be tested,
the display will show the coating thickness.
*If the display shows"--",it means that the tester does not detect any iron,
steel or aluminum substrate.
*If the display shows"OL",it means that the coating thickness exceeds the
tester's measuring range.
*If the display shows"- 0",it means that the probe has wear or the substrate
under the coatings may be other kind of metal other than iron,steel or
4. When you make measurement,the reading will remain on the display if
you move qucikly the tester away from the surface under test;and you can
press the "TEST"button to delete this reading,the display will show"--"
5. If you do not press any button and the tester does not detect the presence
of steel, iron or aluminum for about 30 seconds,the tester will turn off
If you want to turn off the tester manually,press and hold down the "TEST"
button for more than about 1 second.